Clearance Male Tri-tab Vest Skirt
Our Proguard clearance aprons are the perfect option for those who need their radiation protection apparel fast and affordable. So why are these items clearanced? Simple. We make mistakes (sometimes). Those mistakes are usually just an incorrect fabric color, core material or size for an order. No foul play here! Most of these aprons have never even left the facility. So you get a brand new, Proguard quality apron for just a fraction of the price. Plus, we’re not really into the “wasting” thing, so we love seeing our apron misfits find their forever homes. Sounds like a win-win to us.
We update our clearance list regularly, so if you didn’t see anything suited to you, check back another time.
If you’re looking for something more custom, just visit our Proguard Aprons page. Please Note, all clearance apron sales are final.
- Single Front Panel eliminates excess overlapping panels, achieving a 10% weight reduction
- Elastic back offers excellent lumbar support by shifting weight from shoulders to hips
- Tri-tab closure ensures a proper fit and easy wearability
- Non-slip backing on skirt interior to reduce slippage and improve fit
- Premium foam padding in shoulders for extra comfort
- Comes with apron pocket
Made in the USA
- Apron Style: Tri-tab Vest Skirt Apron – Full Protection
- Gender + Size: Male – Small, Medium, Large and X-Large
- Protection (thickness): 0.35mm (front)/0.25mm (back) LE
- Core Material: Premier Lead Free
- Fabric Colors: Coolguard Black Silk (Exterior), Coolguard Indigo (Interior), Black Binding
- Manufacture Date: 9/17/2018
- Apron Notes: Proguard Logo on pocket; Tethered Thyroid Collar Included
- CE Certified: Yes
Protech makes every effort to accurately record its apron weights. Because each category is averaged, the numbers we provide in the weight chart are subject to a +/- 15% weight variance. Minor length or width adjustments, including heavier fabrics, are not differentiated and instead, averaged together. Please consider this when referencing the weight chart. Weights are measured using at 0.50mm LE core material thickness.
|SIZE||VEST LENGTH||VEST WIDTH||SKIRT LENGTH||SKIRT WIDTH||NANO||PREMIER||PROLITE||LEAD|
|X-Small||22.0″||19.5″||20.0″||17.0″||7.53 lbs||8.54 lbs||9.32 lbs||11.16 lbs|
|Small||22.0″||21.0″||20.0″||18.0″||8.60 lbs||9.75 lbs||10.64 lbs||12.74 lbs|
|Medium||24.0″||22.5″||20.0″||20.0″||9.96 lbs||11.29 lbs||12.32 lbs||14.75 lbs|
|Large||25.0″||24.0″||21.0″||22.0″||12.12 lbs||13.74 lbs||15.00 lbs||17.96 lbs|
|X-Large||25.0″||26.5″||21.0″||24.0″||13.20 lbs||14.96 lbs||16.33 lbs||19.56 lbs|
|2X-Large||25.0″||28.5″||21.0″||26.0″||14.48 lbs||16.42 lbs||17.92 lbs||21.46 lbs|
Intended Use: Lead Garments are worn by healthcare professionals to protect them from the harmful effects of x-ray radiation during medical & other procedures requiring the use of x-ray/fluoroscopy. Lead garments are usually worn in combination with other products to provide complete protection from x-ray. You must ensure that your lead garment fits correctly and is properly worn for effective protection.
EN 61331-1:2014: (EU) Protective devices against diagnostic Medical X-ray radiation. (Determination of attenuation properties of material)
EN 61331-3:2014: (EU) Protective devices against diagnostic medical X-radiation Part 3: Protective clothing, Eyewear and Protective Patient shields.
ASTM 2547-18: (USA) Standard test method for determining the attenuation properties in a primary x-ray beam of materials used against radiation.
Maintenance: Lead Garment fabrics can be cleaned by hand using mild soap diluted with room temperature water. Clorox hydrogen peroxide in diluted form is permissible. PDI Sani-Cloth® AF3 Germicidal Disposable Wipes are approved for use on Protech aprons.
Disposal: Users must dispose of Lead Garments and accessories by complying with local, state and federal (or international) regulations where applicable.
Warranty: Protech’s Clearance Radiation Protection Aprons are not covered by our warranty policy. All clearance sales are final.
There are no reviews yet.