Full Glass Mobile Barrier
For personnel in the examination room that need the highest radiation protection, the full glass mobile barrier provides the ultimate, total-body radiation protection during extended procedures. The larger, lead glass window provides unparalleled 2.00mm LE protection and rests on an opaque lower panel with an equivalent 0.50mm LE.
- 2.00mm LE Glass Window provides superior protection
- Smooth rolling and lockable casters provide great mobility
- Lead Glass and Aluminum/Steel base are made in the USA
- Some assembly required
Made in the USA
Not CE Certified
- Composition (Window): Lead Oxide infused glass
- Composition (Base): Lead Oxide Sheeting, Aluminum, Steel
- Protection (Window): 2.00mm LE
- Protection (Base): 0.50mm LE
- CE Certified: No
- Dimensions (Window): 55″ H x 28″ W
- Dimensions (Full): 72″ H x 28″ W x 28″ D
- Weight: 115 lbs
|Lead Glass Shield||2.00mm LE||99.99%||99.9%||99.6%||99.2%|
Intended Use: Lead Barriers are used by healthcare professionals as a mobile solution to protect them from the harmful effects of x-ray radiation during medical & other procedures requiring the use of x-ray/fluoroscopy. Barriers are usually used in combination with other radiation safety products (such as lead garments) to provide complete protection from x-ray.
EN 61331-1:2014: (EU) Protective devices against diagnostic Medical X-ray radiation. (Determination of attenuation properties of material)
EN 61331-3:2014: (EU) Protective devices against diagnostic medical X-radiation Part 3: Protective clothing, Eyewear and Protective Patient shields.
ASTM 2547-18: (USA) Standard test method for determining the attenuation properties in a primary x-ray beam of materials used against radiation.
Maintenance: Examine barrier before use. If the lead glass window or base becomes pitted, chipped or damaged cease use immediately. Basic cleaning maintenance is required to ensure glass is clear and keep casters free of debris.
Cleaning Leaded Barrier Frame: Barrier frame may be cleaned with any non-Ammonia based or non-chlorine based cleaner. Any neutral based disinfecting product may be used for disinfecting if necessary. Never use Ammonia, Bleach or other Acid or Alkali based cleaners
Cleaning Leaded Glass Window: Never use ammonia or chlorine-based cleaners. Window is made with high lead oxide content glass; the surface is susceptible to scratches and attack by acids and alkalis. Use only water, non-abrasive cleaning agents and always wipe dry with a soft cloth. DO NOT allow water to remain or dry as this can discolor/damage the glass.
Disposal: This barrier contains lead in both the upper and lower panels. Users must dispose of Barriers and accessories by complying with local, state and federal (or international) regulations where applicable.
Warranty: Protech’s Leaded Barriers are warranted to be free of defects in materials and workmanship to the original purchaser for one year. If a defect appears, please contact us or an authorized reseller to begin the RMA process.
The warranty is void when the product has been tampered with, when repairs or attempted repairs have been made by unauthorized persons, or when the item has been subject to misuse, abuse or damage in transit. For more information on Returns, Repairs and Warranty, please visit our Returns and Repairs page.
Recently Viewed Products...
Proguard RR Gloves$170.00 – $220.00
Biliary Split Access Drape$296.00
Angiography Drape$229.00 – $296.00
There are no reviews yet.