The angio mitt is a flexible mitt used to shield the hand, thus reducing exposure under direct x-ray. This mitt can be used on either the right or left hand and is mainly used in positioning.
- One size fits all
- Not hand-specific, can be used for left or right hand
Made in the USA
- Core Material Options: Lead, Ultralite, Prolite or Premier
- Thickness: 0.50mm LE
- Customizations: Embroidery and logo
- CE Certified: Yes
- Fabric Options: All fabric types (Click here to view)
- Sizing: One size fits all
|CORE MATERIAL||LEAD EQUIVALENCE||70kVp||90kVp||110kVp||130kVp|
Intended Use: Lead Garments are worn by healthcare professionals to protect them from the harmful effects of x-ray radiation during medical & other procedures requiring the use of x-ray/fluoroscopy. Lead garments are usually worn in combination with other products to provide complete protection from x-ray. You must ensure that your lead garment fits correctly and is properly worn for effective protection.
EN 61331-1:2014: (EU) Protective devices against diagnostic Medical X-ray radiation. (Determination of attenuation properties of material)
EN 61331-3:2014: (EU) Protective devices against diagnostic medical X-radiation Part 3: Protective clothing, Eyewear and Protective Patient shields.
ASTM 2547-18: (USA) Standard test method for determining the attenuation properties in a primary x-ray beam of materials used against radiation.
Maintenance: Lead Garment fabrics can be cleaned by hand using mild soap diluted with room temperature water. Clorox hydrogen peroxide in diluted form is permissible. PDI Sani-Cloth® AF3 Germicidal Disposable Wipes are approved for use on Protech aprons.
Disposal: Users must dispose of Lead Garments and accessories by complying with local, state and federal (or international) regulations where applicable.
Warranty: Protech’s Lead Garments are warranted to be free of defects in materials and workmanship to the original purchaser for two years. If a defect appears, please contact us or an authorized reseller to begin the RMA process. You can also go directly to our Return Merchandise Form and begin the return process.
The warranty is void when the product has been tampered with, when repairs or attempted repairs have been made by unauthorized persons, or when the item has been subject to misuse, abuse or damage in transit. For more information on Returns, Repairs and Warranty, please visit our Shipping, Warranty and Returns page.